Dansensor MAP Check 3 Pressure Online Headspace Gas Analyser
Continuous Monitoring of Gas Output from Gas Mixers or Within Buffer Tanks
Specifically for the continuous monitoring of a gas mix from a pressured source, including from buffer tanks and gas mixers. There are 3 versions available:
Monitor % O2 only – reports % O2 & % Balance.
Monitor % CO2 only – reports % CO2 & % Balance.
Monitor both % O2 + % CO2 – reports % O2 & % CO2 & % Balance
Particularly useful in monitoring the consistency of the gas mix produced by traditional, old technology gas mixers, from which the gas mix fractions can change repeatedly, with changes in the gas input pressures. The test data can be automatically downloaded to Paperless QA software.
Process The Dansensor® MAP Check 3 Pressure continuously draws a gas sample, controls the pressure of the gas sample and passes it through a Dansensor Zirconia O2 sensor and IR CO2 sensor.
A continuous stream of realtime test results are compared against the target values for that specific mix. Alarms are activated if the test results are out of specification. A current, or voltage signal and 24VDC logic for stop/start can also be sent to stop/start the process. The data can also be exported in realtime via a LAN or RS232 ports.
|Technical Specifications||O2 Sensor||CO2 Sensor|
|Key Features||Fastest and most accurate oxygen sensor 0-100% range||Temperature controlled dual beam infrared carbon dioxide sensor 0-100% range|
|Accuracy||±0.01% absolute range below 1% O2 ±1% relative in range above 1% O2||±0.5% absolute ±1.5% relative to reading|
|Heating Time||Heating Time: 10 minutes||Heating Time: 8 minutes|
|Models||Oxygen only, carbon dioxide only, or combined oxygen and carbon dioxide|
|Connections||x2 RS232C, LAN 10/100 Mbit (Modbus TCP), USB, current or voltage output, 24 VDC logic for start/stop process of alarms|
|Power Supply||103-132 / 207-246 VAC (auto ranging), 47-63Hz|
|Dimensions||192 x 230 x 375 mm (H x W x D)|
|Weight||8.5 - 11.5 kg (depending on model)|
|Compliance||CE and Food Safe|
|Gas Inlet Pressure||2 to 10 bar (30 - 145 psi)|
|Gas Media||Dry, clean and non-corrosive gasses|
Further Benefits and Options
To streamline installation, Tendring pacific offers:
- On-site installation and training by our installation engineers
- IP 54 Stainless Steel Enclosure for use in both wet and dusty environments
- Warranty Upgrade
- Annual Service Contracts and Preventative Maintenance Contracts
- Plantstats QC Pro, paperless QA software for data collection with email/SMS alerts
- Dansensor® MAP Mix Provectus – for total control
For more information, please call our sales team on 01223 492083 or e-mail firstname.lastname@example.org